The 256 pneumatic indicator is used to read pneumatic piezometers and settlement cells.
It features a precision flowmeter, a return-flow indicator, convenient controls, a large internal tank, and an analog or digital pressure gauge.
Before leaving for the site, check that the indicator’s tank is charged with gas. The built-in tank holds more than 180 liters of gas (6.25 SCF), which is sufficient for busy reading schedules. On site, set the pressure regulator. Proper regulation results in lower gas consumption and prevents damage to the pressure gauge. Connect transducer tubing to the indicator using the included jumper tubing.
Turn on the gas to activate the transducer. Gas flows down through the tubing to the transducer. Wait for the return-flow indicator to show a return flow of gas, then shut off the gas or use the precision flowrate valve and flowmeter to slow the flow of gas. Wait for the pressure reading on the main gauge to stabilize, then write down the reading. Take a second reading for verification, then disconnect and move to the next transducer.
- Choice of Pressure Gauges
Both digital and analog gauges are available.
- Precision Flowmeter
Use the precision flowmeter when reading with flow or when pressuring long lines.
- High Quality Components
The 256 Indicator is built for hard and long use. It employs the best quality tube fittings, tubing, valves, gauges, and tank.
Twin Tube Indicator Includes
- Pressure gauge
- Precision flowmeter
- Filler hose
- Twin-tube jumper
Please specify range and units for pressure gauge. Indicators shipped by air have empty tank.
|SI51425601||With 0.25% Analog Gauge|
|SI51425602||With 0.05% Digital Gauge